LT Syringe Pump
We put the best materials and designs together using manufacturing processes to ensure optimum alignment of the drive, seal and syringe barrel. The end result is a high value syringe pump that is “elegant in its simplicity.” The LT syringe pump has all the accuracy and precision characteristics of ordinary syringe pumps without syringe replacement schedules and user accessibility concerns. You enjoy longer instrument “up time” and the freedom to design instruments without consuming valuable working area.
Every aspect of the LT syringe pump has been specifically designed to assure trouble free operation for the life of the syringe pump.
- NO calibration required
- NO high fluid pulsation
- NO leaking bellows or diaphragms
- NO ceramics that seize when dry
- NO consumable syringes
Along with a full range of syringe sizes, there are valve, encoder and electronic options to meet almost every instrument application and budget.
|0.150 μL / full step||0.300 μL / full step||0.600 μL / full step||1.500 μL / full step||3.000 μL / full step|
|PPS, Borosilicate Glass, UHMWPE, Viton®|
Imprecision (full stroke)
|< 0.05% CV|
Innaccuracy (full stroke)
2.12 in. [53.85 mm]
5.90 in. [149.86 mm]
2.62 in. [66.55 mm]
.0004 in. [0.01 mm] per full step
Limit Sensor Data
Optoschmitt (Honeywell HOA6990)
15 mA typical
* Pump should typically be driven with a 24 VDC bipolar chopper drive, ½ step, at rated current. Higher voltage may be required at higher step rates.
LT-Series Pump Interface Details
Dimensions/Features noted are typical for any size, LT-Series Pump (5.0 mL shown)
Please contact your TriContinent Sales Representative for current pricing.
LT Syringe Pump with No Electronics and No Valve
|0386||LT Pump, 250 μL|
|0387||LT Pump, 500 μL|
|0388||LT Pump, 1.0 mL|
|0389||LT Pump, 2.5 mL|
|0390||LT Pump, 5.0 mL|
M-Series Controller PCBA
|0956||Controller/Driver Assembly (.5A Max)|
|0959||RS232/485 Converter Kit|
|0960||USB/485 Converter Kit|
Click on file name below to download:
LT Syringe Pump Datasheet (.pdf)
LT-Series Pump 250 μL Model (.igs) 1.1 MB
LT-Series Pump 250 μL eDrawing 154 KB
LT-Series Pump 500 μL Model (.igs) 1.1 MB
LT-Series Pump 500 μL eDrawing 169 KB
LT-Series Pump 1000 μL Model (.igs) 1.1MB
LT-Series Pump 1000 μL eDrawing 194 KB
LT-Series Pump 2500 μL Model 1.1 KB
LT-Series Pump 2500 μL eDrawing 224 KB
LT-Series Pump 5000 μL Model 1.1 MB
LT-Series Pump 5000 μL eDrawing 206 KB
LT-Series Warranty Information
TriContinent operates an ISO 13485 registered manufacturing facility that delivers the highest quality products. LT-Series Pumps are general laboratory components not subject to FDA regulatory approval. TriContinent uses the highest quality components with applicable safety markings to meet the needs of system designers. LT-Series Pumps do not carry independent CE markings as they are components designed for use within larger systems which are subject to independent testing and certification.
We stand behind the LT-Series Pumps with a 1-year warranty. During the warranty period, should a LT-Series Pump fail, call us with an explanation of malfunction and TriContinent will issue a Return Material Authorization (RMA) number for repair or replacement at our discretion, at no charge.
It is the responsibility of the customer to determine the suitability of an application and the material compatibility of our product with your application.
When returning a product for repair or replacement, please contact TriContinent Customer Service to receive an RMA number. When calling please be sure to have the following information available:
Explanation of Malfunction
TriContinent will not accept the return of any product with out prior approval. TriContinent reserves the right to refuse a return of product that has been used with infectious microbiological or radioactive substances or any other materials that may be deemed hazardous.
The RMA number should be referenced on the outside of the shipping container when returning to TriContinent.
The above warranty shall not apply for defective products resulting from:
Improper or Inadequate maintenance by the customer
Unauthorized maintenance or service
Operation outside of the environmental specifications for the product
Instruments returned without proper packaging
There is no warranty expressed for the following components:
Note: For equipment damaged in shipping, please contact the shipper directly.