MC6000 Syringe Pump
The TriContinent MC6000 is a multi-channel syringe pump designed for multiple probe liquid handling applications. This pump introduces an industry exclusive feature of rotary shear valves for optimum liquid handling performance. Our long-life valves were specifically designed and manufactured by TriContinent for multi-channel syringe pump applications to completely eliminate inherent performance limitations (clogging, pumping and leaking) associated with solenoid valves.
Coupled with the ability to use long-life UHMWPE seals, the MC6000 Multi-channel Syringe Pump surpasses all other products of this type in performance, life, reliability and maintenance requirements.
- Long-life rotary shear valves (no solenoid valve limitations)
- Integrated bypass capability
- Full quadrature encoder for absolute syringe positioning
- Available in 4, 6 or 8 channel syringe configurations
- Full stroke resolutions from 6000 to 48000 steps
|Drive Design||Stepper motor/leadscrew with 30 mm travel.|
|Stroke Speed||1.2 seconds to 100 minutes per stroke
(800 minutes in microstep mode).
|Step Resolution||Standard mode 6,000 steps
Hi-Res mode 48,000 steps
|Syringe Design||30 mm stroke glass syringe with 1/4-28 fitting
PTFE seals (long life UHMWPE seals available upon request)
|Fluid Contact Materials||Syringe - Borosilicate glass (seal) PTFE or UHMWPE
Valve - Peek (body) and UHMWPE (rotary plug)
|Imprecision (full stroke) at 50 μL to 100 μL||< 0.1% CV|
|Imprecision (full stroke) at 250 μL to 12.5 mL||< 0.5% CV|
|Innaccuracy (full stroke)||< 1% syringe volume|
|Power Requirement||24 VDC + 10%|
|Power Rating||< 1.5 Amp (peak)|
Envelope and Mounting Drawings
Note: Dimensions in inches [mm] unless otherwise specified.
|No. of Channels||Width||X-Mounting|
|4||5.12 in. [130 mm]||4.74 in. [120.4 mm]|
|6||6.43 in. [163.3 mm]||6.03in. [153.2 mm]|
|8||7.72 in. [196 mm]||7.32 in. [186 mm]|
Please contact your TriContinent Sales Representative for current pricing.
Below pricing does not include syringes
|10154||8-Channel with 120° 3-port Valve PEEK/UHMWPE|
|10155||6-Channel with 120° 3-port Valve PEEK/UHMWPE|
|10153||120° 3-port, ¼ - 28, PEEK/UHMWPE|
|10080||50 μL PTFE|
|10081||100 μL PTFE|
|10082||250 μL PTFE|
|10083||500 μL PTFE|
|10084||1 mL PTFE|
|10085||2.5 mL PTFE|
|10086||5 mL PTFE|
*Long-life UHMWPE seals available upon request
|10171||Cable, RS232 to Pump|
|8246-71||Cable, Interface Converter to Pump|
MC-Series Warranty Information
TriContinent operates an ISO 13485 registered manufacturing facility that delivers the highest quality products. MC-Series Pumps are general laboratory components not subject to FDA regulatory approval. TriContinent uses the highest quality components with applicable safety markings to meet the needs of system designers. MC-Series Pumps do not carry independent CE markings as they are components designed for use within larger systems which are subject to independent testing and certification.
We stand behind the MC-Series Pumps with a 1-year warranty. During the warranty period, should a MC-Series Pump fail, call us with an explanation of malfunction and TriContinent will issue a Return Material Authorization (RMA) number for repair or replacement at our discretion, at no charge.
It is the responsibility of the customer to determine the suitability of an application and the material compatibility of our product with your application.
When returning a product for repair or replacement, please contact TriContinent Customer Service to receive an RMA number. When calling please be sure to have the following information available :
Explanation of Malfunction
TriContinent will not accept the return of any product with out prior approval. TriContinent reserves the right to refuse a return of product that has been used with infectious microbiological or radioactive substances or any other materials that may be deemed hazardous.
The RMA number should be referenced on the outside of the shipping container when returning to TriContinent.
The above warranty shall not apply for defective products resulting from:
Improper or Inadequate maintenance by the customer
Unauthorized maintenance or service
Operation outside of the environmental specifications for the product
Instruments returned without proper packaging
There is no warranty expressed for the following components:
Note: For equipment damaged in shipping, please contact the shipper directly.